On October 28, 2015, Quest Medical, Inc. initiated a nationwide recall of Myocardial Protection System (MPS) Delivery Sets, Models 5001102, 5001102-AS, and 7001102 of specified lots. The product(s) have been found to intermittently exhibit a seal failure during use, which potentially could result in patient blood loss. There are no other recalls related to this product.
Consumers who have the applicable lots of the MPS Delivery Set(s) should cease use of the affected lots and contact Quest Medical, Inc. to return the product and request replacement. Additionally, consignees are being notified via written correspondence.
Recalled Product(s) were manufactured from May 2015 to September 2015 and distributed from June 2015 to September 2015.
SEE ATTACHED PDF
The affected product lot information can be identified by product labeling on shipper and individual sterile trays.
Quest Medical, Inc. voluntarily recalled the applicable lots of MPS Delivery Sets after becoming aware of complaints alleging patient blood loss from the device during use. Quest Medical, Inc. has notified the FDA of this action.
The identified lots of MPS Delivery sets have shown a possible seal failure along the blood source channel of main pump cassette, resulting in blood loss from the bypass circuit and interruption of cardioplegia solution delivery. The firm has received twenty (20) complaints alleging this seal failure which have resulted in 16 instances of patient blood loss during surgery. There have been no reports of patient injuries as a result of the alleged issue to date.
The firm has reported these events via the FDA MEDWATCH program.
Quest Medical, Inc. is notifying its distributors and customers by certified letter/return receipt and is arranging for return and replacement of all recalled product(s).
Consumers with questions may contact the company via telephone at 1-800-627-0226 Monday through Friday between the hours of 9am and 5pm (CT). Consumer may also contact the company via e-mail at [email protected].