This is a new Section. We are trying to develop a database of common or not-so-common problems encountered by perfusionists during CPB.
Please join us by clicking the Link Above if you have had any sort of unusual event you would like to share. Please email details if you feel that your story will help other perfusionists troubleshoot similar issues while on bypass…
To View all submitted cases relating to Perfusion Problem Solving: Click Here
or click the “TroubleShooting” Link on the Side Menu Bar.
International CQI: USA: Cardioplegia Line Kinked
Double checked my ECC prior to a bypass run- had high line pressures when recirculating Quest MPS Cardioplegia delivery system.
Identified and fixed an obvious kink in the outlet line.
Have seen this many times before.
How Was The Problem Identified?
High line pressures when recirculating cardioplegia
What Steps Were Taken ?
- Tried to reposition line
- Placed a supporting splint to eliminate kink Issue Resolved
What Clues Were Missed?
None: Everything was double and triple checked.
This is not the first time I have seen this with this particular delivery system. The kink occurs in the same spot every time.
- No issues.
Want to Share a Problem or Unique Perfusion Issue?
email: [email protected]
(Click Image to View Summary of Glitches)