Click Image Above to Add a “Glitch” to the Perfusion Database …
Editors Note: This is a new Section. We are trying to develop a database of common or not-so-common problems encountered by perfusionists during CPB.
Please join us by clicking the Link Above if you have had any sort of unusual event you would like to share. Please email details if you feel that your story will help other perfusionists troubleshoot similar issues while on bypass…
To View all submitted cases relating to Perfusion Problem Solving: Click Here
or click the “TroubleShooting” Link on the Top menu Bar.
I recently encountered a problem with my cardioplegia line depriming following my initial arresting dose.
It was during the middle of the first distal anastamosis, and I took the clamp off of my cardioplegia line to reposition it. It immediately filled with air. I was shocked because the line was still occluded by the delivery system, so I thought perhaps my aortic root vent was on too high and had created a strong enough negative pressure to pull air out of solution (blood).
As it turned out, my luerlock connection was off by a pinprick- and just loose enough to pull air (negative pressure from vent line) through it when I momentarily moved my clamp.
I re-tightened the connection and I informed the surgeon. We reprimed the line by flushing the air out through the root vent.
Thank you for allowing me to share this event.
How Was The Problem Identified?
- Watched the line deprime.
What Steps Were Taken ?
- Lines reconnected
- De-aired via AO root vent Issue Resolved
What Clues Were Missed?
- None: Everything was double and triple checked.
- First time I ever saw that happen.
- No issues.
Want to Share a Problem or Unique Perfusion Issue?
email: [email protected]
(Click Image to View Summary of Glitches)